The BP Analyzer+ software is easy to use. Follow these simple steps:
1. Download latest BP Analyzer+ software
Download the latest BP Analyzer+ software version by clicking on the link below. Run the application file after download to install BPA+ on the PC.*
2. Connect device to the PC with the USB cable
Connect the blood pressure monitor to the PC using the USB cable. There is no need to switch the device on. 3 horizontal bars will appear blinking on the display of the blood pressure monitor upon successful connection to the PC. Check your USB cable or port if the bars do not show up when connected.
3. Start the data transfer
To transfer measurement data from the blood pressure monitor to the BP Analyzer+ software, open the software while the blood pressure monitor is connected to the PC. When BPA+ detects the blood pressure monitor is connected to the PC, the data transfer button located on the upper right corner in the software user interface appears in blue. Click this button to transfer the data. **
4. Data transfer successful
The message "Measurement data downloaded sucessfully" is displayed upon completion of successful data transfer.
5. Disconnect the device
After completion of data transfer, you can disconnect the blood pressure monitor from the PC. When the blood pressure monitor is disconnected, the data transfer button will turn grey and becomes disabled.
Click here to download the latest BP Analyzer+ software
Make sure the software is being installed while using a Windows administrator account on the PC. The BP Analyzer+ shortcut can be found on the desktop of the PC.
The blood pressure monitor will display "USB" in addition to the three horizontal bars, upon receiving commands (e.g. data transfer) from the BP Analyzer+ software to the PC.
If the option "Clear memory after upload" is turned on, the BP Analyzer+ software will instruct the blood pressure monitor to delete all measurement data in the memory after the data transfer. This option is set to "off" by default.
The date and time setting on the blood pressure monitor are automatically synchronized with the clock settings of the PC during data transfer. Therefore, please also make sure the date and time setting on the PC is correct.
Please make sure the date and time settings of the blood pressure monitor are set, as measurement data without date and time will not be transferred to the BP Analyzer+ software.
The functions and buttons of the blood pressure monitor are disabled during PC connection.
Do not disconnect the USB cable during data transfer.